TO EXPLORE THE UNKNOWN UNIVERSE
C/2022 E3 (ZTF)
Imaging Camera QHY268C
Telescope: Askar 600
Date: January 27, 2023
Location: Aosta, Italy
The QHY1920 is a Scientific CMOS camera with a large 12um pixel size. It is ideal for science research and industry in low light. And it has a relatively high QE in the IR Spectra and can be used in IR applications.
*Please contact QHYCCD about the price.
The QHY1920 has a large pixel size of 12um, which gives it a high sensitivity and an inherent advantage for capturing low light signals. QHY1920 is capable of low light monitoring in various fields and can also be used for live night sky and other science popularization purposes.
|Effective Pixels||2.3 Megapixels|
|Effective Image Area||23mm*14.4mm|
|Full Frame Rate||58FPS@1920*1200 12bit|
|Exposure Time Range||15us – 300sec|
|Anti-Glow Control||Yes. Can be enabled and disabled. Significant Glow Reduction|
|Shutter Type||Electric Rolling Shutter|
|Built-in Image Buffer||128Mbyte DDR2 memory|
|Cooling System||Dual Stage TEC cooler（-32C below ambient）
(Test temperature +20°, exposure >1s)
|Telescope Interface||M54/0.75 female thread on the fast installer/center adjust ring|
|Optic Window Type||AR+AR High Quality Multi-Layer Anti-Reflection Coating|
|Back Focal Length||17.5mm|
All-In-One Pack （Driver, SDK and Software) for WINDOWS supports all QHYCCD USB3.0 devices only except PoleMaster and some discontinued CCD cameras. Please go to https://www.qhyccd.com/download/ and install it.
The camera requires an input voltage between 11V and 13.8V. If the input voltage is too low the camera will stop functioning or it may reboot when the TEC power percent is high, causing a drain on the power. Therefore, please make sure the input voltage arrived to the camera is adequate. 12V is the best but please note that a 12V cable that is very long or a cable with small conductor wire may exhibit enough resistance to cause a voltage drop between the power supply and the camera. The formular is: V(drop) = I * R (cable). It is advised that a very long 12V power cable not be used. It is better to place the 12V AC adapter closer to the camera.
First connect the 12V power supply, then connect the camera to your computer via the USB3.0 cable. Make sure the camera is plugged in before connecting the camera to the computer, otherwise the camera will not be recognized. When you connect the camera for the first time, the system discovers the new device and looks for drivers for it. You can skip the online search step by clicking “Skip obtaining the driver software from Windows Update” and the computer will automatically find the driver locally and install it. If we take the 5IIISeries driver as an example (shown below), after the driver software is successfully installed, you will see QHY5IIISeries_IO in the device manager.
Please note that the input voltage cannot be lower than 11.5v, otherwise the device will be unable to work normally.
Before using software, make sure you have connected the cooling camera to the 12V power supply and connected it to the computer with a USB3.0 data cable. If it’s a planetary/guiding camera, 12V power is not needed.
Note: We recommend 64-bit Software if possible, like SharpCAP x64 , N.I.N.A x64. etc., especially when you’re using 16bit cameras like QHY600.
EZCAP_QT is software developed by QHYCCD. This software has basic capture functions for QHYCCD deep sky cameras.
Run EZCAP_QT. Click “Connect” in Menu -> Camera. If the camera is successfully connected, the title line of EZCAP_QT will display the camera firmware version and the camera ID as shown below.
Click “Temperature Control” in “Camera Settings” to set the temperature of the CMOS sensor. You can turn on “Auto” to set the target temperature. For example, here we set the target temperature to -10C. The temperature of the CMOS sensor will drop quickly to this temperature (approximately 2-3 minutes). If you want to turn off cooling, you can choose Stop. If you just want to set the TEC power but not the temperature. You can select “Manual” and then set the percentage of the TEC power.
You can use the “preview tab” to preview and use the focus tool to focus. Then use the “capture tab” to capture the image.
Launch SharpCap. If the software and drivers mentioned above are installed successfully, the video image will appear automatically about 3 seconds after the software loads. You will also see the frame rate in the lower left corner of the software window as shown below.
If you have already started the SharpCap software before connecting the camera, in order to open the camera, click on the “camera” in the menu bar and then select the device.
Offset adjustment. When you completely block the camera (i.e., like taking a dark frame) you may find that the image is not really zero. Sometimes this will reduce the quality of the image contrast. You can get a better dark field by adjusting the offset. You can confirm this by opening the histogram as indicated in the figure below.
If you want to enter the 16-bit image mode, select the “RAW16” mode.
By selecting the “LX” mode you can expand the exposure setting range and take long exposures.
After cooling devices connected to the 12V power supply, the temperature control circuit will be activated. You can control the CMOS temperature by adjusting the settings in the figure below. Basically, you can control the temperature of CMOS by either adjusting “Cooler Power” or clicking “Auto” and setting “Target Temperature”. You can also see the CMOS temperature at the lower-left corner of the software window.
With ASCOM drivers, you can use the device with many software packages that support the ASCOM standard. We will use Maxim DL below as an example, but a similar procedure is used for The SkyX and other software packages supporting ASCOM.
First make sure you have not only loaded the ASCOM drivers but that you have also downloaded and installed the ASCOM platform from ASCOM. After both the drivers and platform are installed, start MAXIMDL. Follow the instructions shown below to finish the setup. Then Click Connect in and enter the software.
QHYCCD BroadCast WDM Camera is a broadcast driver that supports QHYCCD cameras with video broadcast function, which can meet the needs of customers to send video images to other target software. For example, use sharpcap to connect a WDM-enabled camera, and the sharpcap display video image can be sent to other WDM-supported software for display, which is suitable for video online broadcast applications.
The installation process is over, right-click the computer to find the device manager, and check that the image device name is QHYCCD BroadCast WDM Camera, which means the installation is successful.
HANDYAVI test effect chart:
UFOCAPTURE test renderings:
The CMOS sensor is located in the CMOS chamber. There is a hole in the side of the camera near the front plate that is normally plugged by a screw with an o-ring. If there is moisture in the CMOS chamber that causes the sensor glass to fog, you can connect the silica gel tube to this hole for drying the chamber.
Place an effective silica gel desiccant in the silica tube make sure there is some cotton inside to prevent the silica gel from entering the CMOS chamber.
If you find dust on the CMOS sensor, you can first unscrew the front plate of the cam and then clean the CMOS sensor with a cleaning kit for SLR camera sensors. Because the CMOS sensor has an AR (or AR/IR) coating, you need to be careful when cleaning. This coating can scratch easily so you should not use excessive force when cleaning dust from its surface.
If the ambient humidity is very high, the optical window of the CMOS chamber may have condensation problems. The QHY600 has a built-in heating plate to heat it to prevent fogging. In most cases, it is very effective. However, If fogging still persists, try the following:
1. Avoid directing the camera towards the ground. The density of cold air is greater than the density of hot air. If the camera is facing down, cold air will be more accessible to the glass, causing it to cool down and fog.
2. Increase the temperature of the CMOS sensor. You can increase the temperature of the CMOS sensor slightly to prevent fogging of the glass.
3. Check if the heating plate is working. If the heating plate is not working, the glass will be very easy to fog. Normally, the temperature of the heating plate can reach 65-70 °C in the environment of 25 °C. If it does not reach this heat, it may be because the heating plate is damaged, you can contact us to replace the heating plate.
You should avoid thermal shock during use. Thermal shock refers to the internal stress that the TE cooler has to withstand due to the thermal expansion and contraction when the temperature of the TEC suddenly rises or falls. Thermal shock may shorten the life of the TEC or even damage it.
Therefore, when you start using the TEC to adjust the CMOS temperature, you should gradually increase the TEC power rather than turning the TEC to maximum power. If the power of the TEC is high before disconnecting the power supply, you should also gradually reduce the power of the TEC and then disconnect the power supply.
Now the FAQ part has been intergrated into “QHYCCD Help Center“–Knowledge base
This is QHYCCD Help Center. Here you can:
Submit a Ticket: Describe the issue you met while you’re using them. Our technicans will reply you in 48 hours during working days. You don’t have to check the Ticket update everyday—they can receive email notifications and know if there’s any update.
Knowledge Base: Here lists some tips for using your gears, or solutions to issues that you may met. Help your self!
Support History: Check your ticket’s status.